Images are representations only.
MIST PKIT Programming KitBrand: Intempco Controls, Inc.
The MIST PKIT-1 is used to re-calibrate and re-scale MIST Series sensors. Two methods of calibration are available.
The first, called One Point Calibration, is performed by setting the MIST to a known temperature and adjusting the 4-20 mA output to the corresponding value.
The second method, called Two Point Calibration, is performed by setting the MIST to a known point in the operating range and then recording this point by simply clicking the corresponding button.
The next step is to set the MIST to a second known point in the operating range and record it in the same manner.
Finally, by clicking on the “Process” button, the calibration parameters are automatically calculated and saved in the MIST memory.
The MIST software can also be used to re-scale the MIST sensor output within its operational limits. A choice of many pre-scaled values is offered, as well as a custom range.
Once the range is selected, simply clicking on the “Set” button will automatically re-scale the output of the MIST and save the new values in the MIST memory.Features
- MIST™ calibration software CD-ROM
- Communication module with RS-232 or USB
- 110 VAC Wall adapter (RS-232 only)
- 100 Vac Wall Adapter
- Windows® XP/Vista, Pentium III - 600 MHz or Better, 10 MB of Free Memory, 256 MB of RAM, 1 Serial Port or USB